ASTM Level 3 Surgical Mask
This product is authorized by the FDA as a surgical mask under EUA for use in healthcare settings by HCP as PPE to provide a physical barrier to fluids and particulate materials to prevent HCP exposure to respiratory droplets and large particles during surgical mask shortages resulting from the COVID-19 pandemic. This product has not been FDA cleared or approved.
Lightweight. Easy on & off design with ear loops to secure fit
Comfortable and non-irritating, and not made with natural rubber latex
Pleated design fits wide variety of faces
ASTM Level 3 Performance
Included in box: 50, included in carton: 2000
Full Berry Certificate available
All direct and indirect skin contacting materials are biocompatible per ISO 10993. List of body contacting materials below.
Outdoor Research Surgical Masks have not been FDA cleared or approved. Outdoor Research Surgical Masks are authorized by the FDA only for the duration of the declaration that circumstances exist justifying the authorization of the emergency use of medical devices, including alternative products used as medical devices, during the COVID-19 outbreak, under section 564(b)(1) of the Act, 21 U.S.C. § 360bbb-3(b)(1) unless the authorization is terminated or revoked sooner. They are intended to be used when FDA-cleared surgical masks or FDA cleared or authorized respirators are not available. The Outdoor Research Surgical Mask is a loose-fitting device worn over the user’s nose and mouth. Surgical Masks may be effective in blocking splashes and large-particle droplets; however, because of the loose fit between the surface of the surgical mask and the user’s face, leakage can occur around the edge of the mask when the user inhales. Therefore, a surgical mask may not provide the user with a reliable level of protection from inhaling smaller airborne particles and is not considered respiratory protection. For this reason, surgical masks are not recommended for use in aerosol generating procedures and any clinical conditions where there is significant risk of infection through inhalation exposure. In such clinical conditions, a filtering facepiece respirator (such as an N95 respirator) with a tight fit is recommended to provide a more reliable level of respiratory protection against pathogenic biologic airborne particulates.
Instructions for Use
- Thoroughly inspect the surgical mask prior to use. If any components, including the filter material, the elastic earloops, or the nosepiece are damaged or missing, do not use and dispose of properly. Damage may include missing components, tears or holes in the mask filter media, or other issues which noticeably have a negative effect on performance.
- Immediately leave the contaminated area and seek help if dizziness, difficulty breathing, or other deleterious conditions arise.
- This is a disposable, single use surgical mask. It may be used until it becomes difficult to breathe through, damaged, or is thought to be contaminated in any way. In a healthcare setting, it should be discarded after each examination or procedure. Given the Public Health Emergency PPE shortage, follow current government and facility guidance and regulations.
- A surgical mask is not intended to replace FDA-cleared surgical masks or FDA-cleared or authorized respirators and does not protect against pathogenic biological airborne particulates. This device should not be used in aerosol generating procedures or any clinical conditions where the risk of infection through inhalation exposure is high.
- Pull the earloops to the outside of the mask.
- With the wire nosepiece up, and downward-facing pleats outward, pull the elastic earloops behind the ears. The mask should be secure over the nose and mouth.
- Press the nosepiece so it conforms to the bridge of the nose and pull the mask down to wrap under the chin.
- Without touching the surgical mask filter media, remove the earloops from behind the ears and the mask from the face. Be mindful not to the touch face or mask filter as hands may be contaminated. Dispose in accordance with current government and facility guidance and regulations
Outdoor Research Headquarters - Seattle, Washington
Middle Filter Layer
Coated Steel Wire
All direct and indirect skin contacting materials are biocompatible per ISO 10993.
How Does Outdoor Research Compare?
Outdoor Research Surgical Masks meets ASTM F2100-2019 Level 3 performance.
|Tests Performed||Tests Method||Performance|
|Barrier Tests||Bacterial Filtration Efficiency||ASTM F2101||≥98%|
|Barrier Tests||Particle Filtration Efficiency||ASTM F2299||≥98%|
|Barrier Tests||Fluid Splash Resistance Pressure||ASTM F1862||160 mm Hg|
|Physical Tests||Differential Pressure||EN 14683 Annex C||<6.0 mm H2O/cm2|
|Safety Tests||Flamability||16 CFR Part 1610||Class 1|
|Biocompatibility||Sensitization, Irritation||ISO 10993-10||Non-Irritating and Non-Sensitizing|